|Measuring Range:||0~25m||Power Supply:||24VDC|
|Protection Grade:||Display Instrument IP67, IP67||Operating Temperature:||-40~130℃，-40~250℃|
|Signal Output:||4-20mA, 4-20mA/HART(Two Wire/FOUR Wire)||Process Pressure:||Atmospheric Pressure,-0.1-4MPa, -1~40BAR|
24VDC Magnetostrictive Level Sensor,
Magnetostrictive Level Sensor 316L,
IP67 Magnetic Level Transmitter
NYCZ Cable Type Magnetostrictive Liquid Level Meter Level Transmitter
Magnetostrictive Liquid Level Meter
Magnetostrictive Liquid Level Meter Operating Principle
The principle of magnetostrictive level gauge is shown in figure. The products mainly consist of measuring rod, electronic warehouse and non-contact float on the measuring rod. The rod is equipped with magnetostrictive wire (waveguide wire), which is made of stainless steel tube without magnetic conductivity. It protects the waveguide wire very well.
When the magnetostrictive liquid level meter is working, the pulse generator in the electronic chamber sends out an initial pulse, which, as it travels through the waveguide wire, produces a rotating magnetic field moving along the waveguide wire. When the magnetic field meets the permanent magnetic field in the float sphere, a magnetostrictive effect is produced, resulting in a constant velocity mechanical wave caused by the twisting of the waveguide wire. The mechanical wave propagates in two directions, and the mechanical wave at the end of the rod is absorbed by the damper, and the mechanical wave that returns the direction of the electron bunker is sensed by the pickup mechanism installed in the electronic chamber and converted into the corresponding current pulse. The pick up mechanism calculates the time between the two pulses The height of the measured liquid level can be accurately measured by the difference.
Magnetostrictive Liquid Level Meter Technical Parameter
Probe: 0-50~6000mm Cable: 25m
|Power supply||24VDC ±10%|
|Signal output||4~20mA / Hart / RS485|
|Working current||4~20mA; 20.4mA|
|Load characteristics||when current output, maximum load resistance 600 Q voltage output, maximum load Resistance 2mA|
|Process temperature||-40〜130 °C,-40〜250 °C|
|Temperature effect||±0.5 °C|
|Process pressure||- 0.1 ~ 4.0MPa|
|Response time||0.2 ms|
|Rod material||304/316L (optional)|
|Process connection||Thread, flange, Support mounting (optional)|
|Site display||LCD (optional)|
|Housing material||stainless steel / aluminum (optional)|
|Explosion proof grade||Exia IICT6 (optional)|
※ Note: Products can be customized according to the specific requirements of users!
Magnetostrictive Liquid Level Meter Design Features
Magnetostrictive Liquid Level Meter Installation Requirement
Magnetostrictive level gauge can be used in flammable, explosive and corrosive liquid applications. The electronic chamber of the magnetostrictive liquid level gauge is spattered but not submerged, and the liquid must not be immersed above the hexagonal base. The float ball has an installation direction and the NC-marked hemisphere should be above the liquid level. The installation method is as follows:
Method 1: according to the connecting flange above, the installation hole is processed on the flange cover and directly rotated into the magnetostrictive liquid level meter, as shown in fig. 1;
Method 2: fixing magnetostrictive liquid level meter with sensor support and locking nut, this method is suitable for liquid level measurement of open tank, as shown in figure 2;
Method 3: machining a G2 thread hole on the cover of the tank fbr convenience Place the float, as shown in figure 3;
Method 4: the length of the measuring rod can be adjusted according to the height of the liquid level by using.
Magnetostrictive Liquid Level Meter Electrical Connection
the movable connecting head, as shown in figure 4
◎The output zero position of magnetostrictive liquid level meter is one side at the end of the rod and the full range output position is on the side of electronic warehouse.
◎The shielding line of transmitter must avoid high power supply, RF signal source and other noisy transmission lines.
The cable shield must be kept intact and connected.
Contact Person: Gao