|Flange Mounting Or Thread Mounting
|Display Instrument IP67, IP67
|4-20mA, 4-20mA/HART(Two Wire/FOUR Wire)
IP67 GWR Radar Level Transmitter,
316L GWR Radar Level Transmitter,
IP67 guided radar level sensor
NYRD701 ( Stainless Steel )1.8GHz Guided Wave Radar Level Meter
Technical advantages of guided wave radar level gauge: radar level gauge for liquid, particle and slurry continuous level measurement. The measurement of radar level meter is not affected by medium change, temperature change, inert gas and steam, dust, foam, etc. The accuracy of the radar level meter is 1mm, the range is 30 meters, the high temperature of 250 degrees, 40 kg high pressure, the radar level meter is suitable for the explosion danger area.
Guided wave radar level gauge is a kind of level gauge based on TDR principle. The electromagnetic pulse of the level gauge travels at the speed of light along the cable or probe. Part of the pulse of the level gauge is reflected to form an echo and returns to the pulse transmitter along the same path. The distance between the transmitter and the surface of the measured medium is proportional to the propagation time of the pulse in that time period, and the height of the liquid level is calculated.
Radar Level Meter Technical Parameter
|HART communication protocol
|-1 ~ 40bar
Power: 24VDC (± 10%), ripple voltage: 1Vpp
|temperature -40 ℃ ~ +80 ℃
|Housing protection grade
|Instrumentation power supply and signal output wire connection
Sharing a two-core shielded cable
※ Note: Products can be customized according to the specific requirements of users!
Radar Level Meter Operating Principle
Guided Wave Radar is the kind of measuring instrument that is based on the principle of time travel. The radar wave travels at the speed of light, and its running time can be converted into a level signal by the electronic components. The probe emits high-frequency pulses which spread along the cable, and the pulses are reflected back when they encounter the material surface and received by the receiver inside the meter. The distance signal is then converted to the level signal.
The reflected pulse signal is conducted along the cable back to the electronic circuit of the meter, and the microprocessor processes this signal to identify the echo produced by the microwave pulses at the material surface. The recognition of the correct echo signal is done by the intelligent software. The distance D to the material surface is proportional to the time travel T of the pulse:
where C is the speed of light
As the distance E of the empty tank is known, the level L is:
Set accordingly to the empty tank height E (= the zero point), the full tank height F (= the full scale) and some application parameters. The application parameters will automatically make the instrument adapt to the measurement environment, corresponding to the output of 4 ~ 20mA.
Radar Level Meter Design Features
Radar Level Meter Measurement Distance
The following table lists the relationship between the different categories of the measured medium and the measuring distance.
|－Condensing gas,e.g. N2CO2
|3m(Refers only to the coaxial rod probe)
－Plastic belt particles
－Liquefied petroleum gas,e.g. propane
－Wet stone, ore
- Chlorobenzene, chloroform
- Metal powder
- Aqueous liquids
Contact Person: Lay