|Mounting Method:||Universal Flange, Clamp And Bracket Are Optional||Shell Material:||Stainless Steel|
|Echo Sampling:||100 Times/s||Operating Temperature:||-40~80℃|
|Signal Output:||4...20mA/HART/RS485/Modbus...||Protection Grade:||IP67|
12VDC Radar Level Transmitter,
Radar Level Transmitter 10m,
10m Microwave Level Transmitter
26GHz NYRD-810 Radar Level Meter
The NYRD-810 parabolic antenna radar level gauge is a non-contact radar level gauge that uses the pulse principle. It can be widely used to measure solid material levels, process containers, strong dust, easy crystallization, and condensation.
Radar Level Meter Technical Parameter
|Typical application||Monitoring Solids, dust, particles, etc|
|Antenna material||304 316L (optional)|
|Echo sampling||100 times/s|
|Process connection||Flange or bracket installation|
|Electrical Interface||M20×1. 5, ½ " NPT|
|Supply power||24V DC (optional)|
※ Note: Products can be customized according to the specific requirements of users!
Radar Level Meter Working Principle
Measurement principle The radar level meter transmits a microwave pulse signal with a narrow J and very low energy through an antenna. This pulse signal is transmitted in space at the speed of light and is reflected when it encounters the measured medium. The reflected signal is received by the instrument, and the pulse signal is transmitted and the pulse signal is received. The time interval is proportional to the distance between the reference plane and the surface of the measured medium. The distance between the antenna and the surface of the medium can be measured by measuring the time interval between transmission and reception.
Contact Person: Gao